Provided by: argyll_2.0.1+repack-1_amd64
Read - Read Target Test Chart.
Read Target Test Chart
chartread [-options] outfile -v Verbose mode -c listno Set communication port from the following list (default 1) ** No ports found ** -t Use transmission measurement mode -d Use display measurement mode (white Y relative results) -e Emissive for transparency on a light box -p Measure patch by patch rather than strip -x [lx] Take external values, either L*a*b* (-xl) or XYZ (-xx). -n Don't save spectral information (default saves spectral) -l Save CIE as D50 L*a*b* rather than XYZ -L Save CIE as D50 L*a*b* as well as XYZ -r Resume reading partly read chart -I file.cal Override calibration info from .ti2 in resulting .ti3 -F filter Set filter configuration (if aplicable): n None p Polarising filter 6 D65 u U.V. Cut -N Disable initial calibration of instrument if possible -B Disable auto bi-directional strip recognition -H Use high resolution spectrum mode (if available) -X file.ccmx Apply Colorimeter Correction Matrix -X file.ccss Use Colorimeter Calibration Spectral Samples for calibration -Q observ Choose CIE Observer for CCSS instrument: 1931_2 (def), 1964_10, S&B 1955_2, shaw, J&V 1978_2 -T ratio Modify strip patch consistency tolerance by ratio -S Suppress wrong strip & unexpected value warnings -W n|h|x Override serial port flow control: n = none, h = HW, x = Xon/Xoff -D [level] Print debug diagnostics to stderr outfile Base name for input[ti2]/output[ti3] file